ISO 11608: „Kanülenbasierte Injektionssysteme zur medizinischen Verwendung – Anforderungen und Prüfverfahren“
Desweiteren wird im USP (United States Pharmacopeia) Chapter <1382> und <382> insbesondere auf die Norm ISO 11608-3 referenziert.
Bezogen auf die ISO 11608-3 werden hier die Anforderungen an die funktionalen Eignungsprüfungen zur Losberech-, Gleitkraft und Dichtigkeit beschrieben.
- USP <1382> Dieses Kapitel enthält Informationen und Anleitungen zur Unterstützung bei der Bewertung der funktionalen Eignung von Elastomerkomponenten als Teil von Verpackungs-/Arnzeimittelabgabesystemen, die für parenterale Darreichungsformen vorgesehen sind.
- USP <382> Dieses Kapitel behandelt die Anforderungen an die funktionelle Eignung von Verpackungs-/Arzneimittelabgabesystemen, die für parenterale Darreichungsformen bestimmt sind.
- The most important requirements when testing medical products are the reproducibility of results and the minimization of operator influences. To meet these requirements, ZwickRoell has developed automated systems for testing pen injectors (for example, for insulin). Insulin therapy normally involves the subcutaneous injection of insulin by means of prefilled syringes or insulin pens. These pens are similar to writing pens and are filled with insulin cartridges. Single-use pens are disposed of after the contents of the cartridge are dispensed, whereas reusable pens can be used year after year. For quality assurance tests on insulin pens and carpules, the standardDIN EN ISO 11608-1, DIN EN ISO 11608-2 and DIN EN ISO 11608-3 applies.
- ZwickRoell testing systems and test fixtures also fulfill the requirements of USP (United States Pharmacopeia) Chapter <1382> and <382>.
A testing system consisting of a zwickiLine table-top testing machine with additional torsion drive is used to test insulin pens. This testing system can measure various pen functions, such as the dosage setting, actuation force, glide force, and specified dosage in one continuous process. You can modify and combine the test methods of both test axes to suit your testing needs.
The testing system can perform these tests to check the pen's mechanical function without the drug. Along with force and torque, it measures the activation force with the utmost precision. When testing filled pens, an integrated high-resolution scale is used to measure the dosing accuracy to DIN EN ISO 11608-1.
Performing a test on insulin pens using an automated testing system together with the zwickiLine materials testing machine with integrated torsion drive ensures reliable and cost-effective test results . Only one test fixture is needed to check a variety of insulin pen functions. This system allows you to measure the dosage setting, actuation force, glide force, and specified dosage in one continuous process. You can modify and combine the test methods of both test axes to suit your testing needs. The robotic testing system roboTest R is responsible for automated specimen feeding.
It uses the automation software autoEdition3 to remove insulin pens from the magazine, feed them into the testing machine, and start the test. This minimizes the possibility of inaccurate test results due to operator influence. Our solution makes the test process significantly more efficient thanks to increased specimen throughput. It is also possible to manually test specimens at any time.
testXpert III together with the Expanded Traceability option to FDA 21 CFR Part 11 make it possible to create documentation for the testing process that is complete and tamper proof.
- Ever-increasing demands are placed on software used in the medical and pharmaceutical industries to document the traceability of completed actions.
- With the traceability option, testXpert III enables logging of all actions and changes before, during and after the test, making test results and the documentation traceable and protecting them from manipulation.
- Integrated user management and functions such as electronic records and electronic signature ensure that test results are always protected from tampering.
- Together with the organizational measures and procedure instructions that apply to the individual companies themselves, the requirements of FDA in 21 CFR Part 11 are fulfilled.
- ZwickRoell also offers a qualification service package (DQ/IQ/OQ) for validation support.
- testXpert III logs all test and system related actions and settings and can therefore always answer the question “When does who do what, why and who is responsible?”
Learn more about the testXpert III option